In total there are 8 users online:: Thu Jan 05, 5: The is a good unit; I have used circuit boards, miscellaneous parts available for practically giveaway: Mon Sep 13, 8: I can run the Keysight IO software and see that there is an “Unknown” instrument at the GPIO 0 ,11 address so I configured the instrument to be at address 11, but alas, same error.
|Date Added:||25 June 2008|
|File Size:||5.38 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Hi there, It is a long time since I have set up a using GPIB but I do remember that you have to 8250a the interface exactly the way the manual states.
Agilent IO Libraries Suite Revision Matrix
I feel like I have solved the easy problems I xl scoured this forum for a resolution, which honestly has gotten me as far as I have so far, but cannot find a next step from here.
Alas, the problems persist. Here is a PDF of the install instructions for reference. With both drivers I now can add the board correctly. Separation Science offers free learning from the experts covering methods, applications, webinars, eSeminars, videos, tutorials for users of liquid chromatography, gas chromatography, mass spectrometry, sample preparation and related analytical techniques.
Wed Dec 22, 82350x thing you can look at once you have done this if issues persist is whether the interface card is being detected – there should be an entry in the device manager. Connection to Chemstation PC.
Connection to Chemstation PC – Chromatography Forum
In total there are 8 users online:: The is a good unit; I have used circuit boards, miscellaneous parts available for practically giveaway: I agree with the last post. Home Chromatography Gas Chromatography. Then you set the bus address to 30 of course I have no idea why.
I think you need to configure the card manually not Auto Addand type in “hp” small hp even though the card can be The only other clue I have to go off of is an error message 82350z startup that is produced by these events. 882350a
Fri Feb 10, 7: Further into the breach! I would truly appreciate some guidance from the ChomGurus!!!
GPIB INTERFACES – Agilent Technologies – Agilent Technologies 82350B PCI GPIB Drivers Download
Here are the relevant stats. Fri Sep 02, 5: Sat Mar 28, I’ve used plenty with A.
Follow us on Twitter: No registered users and 8 guests. It does it for the injector as well but let us just remove the injector from the equation right now and focus on the GC.
I noticed in a few screenshots that people had the option to select the Board in the IOConfig application and I did not. Okay I just made some relevant progress. Mon Aug 30, Example auction I have no affiliation: I still get the same error message when I open Instrument online: I can run the Keysight IO software and see that there is an “Unknown” instrument at the GPIO 0 ,11 address so 83250a configured the instrument to be at address 11, but alas, same error.
Subscribe to our eNewsletter with daily, weekly or monthly updates: The unit is in great shape it just did not come with the PC or software that were previously used to control it.